The conditions (humidity, pressure, temperature, etc.) of the medium surrounding the load cell.
The temperature of the medium surrounding the load cell.
Angular Load Eccentric--角载荷偏心：
A load applied eccentric with the primary axis at the point of application and at some angle with respect to the primary axis.
Angular Load Concentric--角载荷同心：
A load applied concentric with the Primary axis at the point of application and at some angle with respect to the Primary axis.
A load applied along or parallel to and concentric with the primary axis.
The comparison of load cell outputs against standard test loads.
A record (graph) of the comparison of the load cell outputs against standard test loads.
(Non linearity and Hysteresis) The maximum deviation from the straight line drawn between original no--load and rated load outputs expressed as percentage of the ratedoutput and measured on both increasing and decreasing loads.
The utilization of supplementary devices, materials, or process to minimize known sources of error.
The change in load cell output occurring with time while under load and with all environmental conditions and other variables remaining constant.
The change in no--load output occurring with time after removal of A load which had been applied for a specific period of time. Usually measured over a specific time period immediately following removal of rated load and expressed as a percent of rated output over a specific period of time.
The change of length along the primary axis of the load cell between no--load and rated load conditions.
A random change in output under constant load conditions.
Any load applied parallel but not concentric with the primary axis.
The algebraic difference between the indicated and true value of the load being measured.
The voltage or current applied to the input terminals of the load Cell.
The range of frequencies over which the load cell output will follow the sinusoidally varying mechanical input within specified Limits.
The signal (voltage, current, pressure, etc.) produced by the load cell. Where the output is directly proportional to excitation, the signal must be expressed in terms of volts per volt, per ampere, etc., of excitation.
The algebraic difference between the outputs at no--load an at rated load.
Rated Capacity (Rated Load)--额定容量(额定负载)：
The maximum axial load the load cell is designed to measure within its specifications.
A force measuring device whose characteristics are precisely known in relation to a primary standard.
The smallest change in mechanical input which produces a change in the output signal.
The ratio of the change in output to the mechanical input.